MAC1256 – RT 250 DIGITAL
TEAM RADARTERAPIA RALP. EMISSION MODE continuous and pulsed.
- 250W Max Power 1600W continuous mode and pulsed mode.
- Display 240×64 DPI
- 20 stored programs.
- 10 programs to memorize.
- Armed with a circular antenna arm and orthostatic AC model
- 1 Power cord
- 1 Manual
- Two 6.3-T Fuses
- A frequency output cable
- 1 Antenna parts of this series AC
- 1 Smart Card
- A pantograph arm
Medical use of microwaves is mainly based on the fact that they are selectively absorbed by tissues with high water content and thus allow selective heating of some areas such as muscles, while the penetration and absorption bone level is minimal.
Electromagnetic waves, once they come into contact with organic tissue, transform the radiant energy into heat (thermal energy) with a different degree of penetration predominantly restricted to muscle and periarticular tissues (skin and subcutaneous tissue, tendons, membrane, muscle strips , synovium).
The family therapy equipment or radarterapia mircroondas is composed of two models that differ only by the mode of power delivery.
The core of the team is represented by a device consisting of a cavity oscillator capable of generating nuea the electrical frequency of a sinusoidal alternating current to about 2.5 Ghz. and operate either in continuous or pulsed mode: 25% – 50% – 75%.
The availability of pulsed operation as an alternative to continuous treatment, and permits also with relatively long applications (where necessary) without causing the patient feelings of discomfort due to excessive furnishing of heat.
The use of smart card allows patient crar personalized cards.
A fast convenient encoder to select current or dispense protocol by accessing a library of over 100 programs or create and store up to 300 protocols.
|Specifications||2500CP||2500C||Rt 250 Digital|
|Power supply||230 VAC ,50-60 Hz ± 10%||230 VAC ,50-60 Hz ± 10%||230 VAC ,50-60 Hz ± 10%|
|Power max absorbed by the network||650 VA||650 VA||650 VA|
|Double fuse type network ritardato di||6.3AT||6.3AT||6.3AT|
|Backlit LCD display for viewing and control of operating parameters||Grafico320x240 DPI||Grafico320x240 DPI||Grafico240x64 DPI|
|Treatment Time Programmable||1-30 minutes||1-30 minutes||1-30 minutes|
|Peak Power 50 Ohm pulsátilsu adattato carico||1600 W||1600 W|
|Peak power continues to carico adattato 50 Ohm||250 W||250 W||250 W|
|Insulation class||I type BF||I type BF||I type BF|
|Hazard Class (93/42/EEC)||IIB||IIB||IIB|
|Degree of protection of liquids||IPXO||IPXO||IPXO|
|Protocols memorized||100 -? 100||20|
|Stored in Smart Card Protocols||200 -? 200|
|Protocols storable in internal memory||200 -? 200||10|
|Alarm in case of overheating||●||●||●|
|Diagnostic test to verify the operation of the machine||●||●|
|I use low-noise fans||●||●||●|
|Weight||40 Kg||40 Kg||40 Kg|
|Size||385x865x310 mm||385x865x310 mm||385x865x310 mm|